Next: Near edge structure of Up: High-Energy Quantum Efficiency from Previous: A typical example and
The technique described above was applied to all frontside illuminated flight devices. Depletion depths were extracted from the Fe55 data sets taken at the HIREFS spectrometer chamber. These data have lower flux rate than other chambers. Results are summarized in Table 4.19.
|Device||Location||Depletion Depth ()||Remarks|
| PI = +2,+10V
| PI = -5,+5V
(reduced dark current)