Sun Aug 8 17:17:30 1999 : obs_anal.pro started
Filename = /nfs/spectra/d0/mta/letg_hrc_capella.p4a.fits Output_prefix = letg_hrc_capella.p4a Output dir. = /spectra/d0/Obs_out Grating = LETG cd_width = 5.00000 mm cd_offset = 0.00000 mm Detector = HRC-S Spatial coord = DET (FLIPY is not set) ASPECT keyword set
Source of data: /nfs/spectra/d0/mta/letg_hrc_capella.p4a.fits Reading FITS file: /nfs/spectra/d0/mta/letg_hrc_capella.p4a.fits Tag names are: TIME CHIP_ID CHIPX CHIPY TDETX TDETY DETX DETY X Y PHA SHELL ORDER V U ZCOS YCOS XCOS ZPOS YPOS XPOS MARX_ENERGY * No ENERGY available. Creating ftime from TIME Total Events = 23868 Start time = 35647192. seconds Interval duration = 4999.7830 seconds Ave Event Rate = 4.7738071 events/second
Spatial coordinates are : DETX, -DETY pixel size = 6.4268 microns Processing method = aDET
Keeping TIME-in-range events, 100.000 % (from 0.0000000 to 4999.7830) Start time = 35647192. seconds Interval duration = 4999.7830 seconds HRC: Keeping central stripe Y events, 100.000 % No STATUS selection No QUALCODE selection No GRADE selection No ENERGY selection Keeping HRC PHA selected events, 98.9316 % (from 1.00000 to 250.000 )
Selected Events = 23613 Start time = 35647192. seconds Interval duration = 4999.7830 seconds Ave Event Rate = 4.7228049 events/second Found no big ( > 15.0000 s) gaps in event data.
Average DETX,-DETY gives x,y = 32823.9, -32766.8 Average DETX,-DETY gives x,y = 32774.4, -32767.2 Average DETX,-DETY gives x,y = 32769.8, -32767.5
==> MTA Monitor Task 5.2 <==
Zero-order Location DETX,-DETY ave-median = 32770.3, -32767.0
Events in Zero-order Region = 10014 Ave. Event Rate in Zero-order Region = 2.0028869 events/seconds DETX,-DETY RMS Radius = 91.3436 pixels Number of DETX LRF pixels GT 10% of peak = 130 Number of DETX LRF pixels GT 50% of peak = 25
Aspect solution from DETX,-DETY data... Calculating DETX,-DETY averages every 60.0000 seconds. Apodizing during the first and last 833.29717 seconds of data. Requires 83 bins ... Min / Max number of events per time bin = 92 / 154
Making aspect corrected coordinates, AX,AY ... Before aspect correction: DETX,-DETY RMS Radius = 91.3436 pixels Number of DETX LRF width at 10% of peak = 130 Aspect correction yields (including apodized events): AX,AY RMS Radius = 72.518472 pixels Width of AX LRF at 10% of peak = 14.6 pixels
Total Events = 15660 Start time = 35648017. seconds Interval duration = 3333.0537 seconds Ave Event Rate = 4.6983941 events/second The spatial coordinates to be used for grating spectra are: AX, AY = aDETX, a-DETY AX,AY RMS Radius = 70.758721 pixels Width of AX LRF at 10% of peak = 75.2 microns Width of AX LRF at 50% of peak = 30.8 microns Gaussian fit FWHM = 44.0 microns FWHM error = 0.4 microns
Sky X,Y (Level 1) Coordinates are present... showing them too. X,Y RMS Radius = 70.727897 pixels Width of LRF at 10% of peak = 54.0 microns Width of LRF at 50% of peak = 21.2 microns Gaussian fit FWHM = 29.7 microns FWHM error = 0.3 microns
==> MTA Monitor Task 5.5 <==
Selecting events in the grating parts... cd_width = 5.00000 mm cd_offset = 0.00000 mm LEG_minus number of events = 4326 LEG_minus slope is -0.00017988601, --> -0.010306709 degrees on detector LEG_plus number of events = 4422 LEG_plus slope is 0.00043239015, --> 0.024774129 degrees on detector LEG_all number of events = 8748 LEG_all slope is 0.00015794014, --> 0.0090493036 degrees on detector
Rowland spacing used is 8634.00 mm ACIS ENERGY not available: keeping all orders Constant d-Lambda binning from 0.0500000 to 10.0000 keV E/dE of one bin width at 1 keV = 1666.06
==> MTA Monitor Task 5.7 <==
Smoothing the spectra by E/dE ~ 40.5603
Creating letg_hrc_capella.p4a_aDET_LEGmAll_hist.rdb ... Creating ISIS format: letg_hrc_capella.p4a_aDET_LEGmAll_isis.dat ... Creating letg_hrc_capella.p4a_aDET_LEGpAll_hist.rdb ... Creating ISIS format: letg_hrc_capella.p4a_aDET_LEGpAll_isis.dat ...
No Level 1.5 columns found in the file.
No Level 1.5 columns - cannot create L1.5 histograms.
Sun Aug 8 17:18:14 1999 : obs_anal.pro finished