|back: H-HAS-EA-8.002||back to Analysis Table||next: H-HAS-EA-8.004|
XRCF Data (upper: all iterations) and MARX Simulation (lower: first iteration only)
Energy scan: 1.400, 1.430, 1.460, 1.490, 1.520, 1.550, 1.580, 1.610, 1.640, 1.670, 1.700H-HAS-EA-8.003 Notes
DCM energy scan.
Need to analyze for effective areas and order ratios.
Defocus set to 35 mm (CMDB says it was 40.)
This is the data set where Herman caught the differences in the +/- diffracted images ring widths/ellipticities. It is due to the DCM energy varying across the HRMA entrance aperture!