Next: Version History Up: Overview of calibration Previous: Filter calibration
The ACIS team collected calibration data utilizing the HRMA telescope during three data collection intervals. Following the XRCF nomenclature these occured during Phases F, G and H. During Phases F and G the HRMA focussed X-rays were collected on two flight-like CCDs (one front side illuminated (FI), and one back side illuminated (BI)) in a camera utilizing electronics very similar to the MIT CSR lab cameras. This camera, named the ACIS-2C, served as a proxy to enable data collection while the ACIS flight instrument finished the final stages of certification and thermal vacuum testing.
The ACIS-2C camera, using chips of identical design and fabricated in the same lots at Lincoln Lab as the flight ACIS chips, has differing processing electronics and does not hold the chips in the same orientation as the flight unit. Thus these ACIS-2C data are useful for understanding the chip level interaction between the mirror effects and the CCDs, but do not fully calibrate the effect of the flight electronic processing and flight array orientation effects.
Measurements were collected using the flight ACIS instrument and the HRMA at XRCF for eight days during Phase H (see Table 2.1 for the times of ACIS data collection). These data are the most directly relevant data as they use the flight mirrors and detectors, but must still be adjusted for the effects of finite conjugate distance, gravitational loading on the mirrors and a slightly differing CCD temperature operating point than planned for flight (-115 C vs. -120 C).
Following the departure of the HRMA mirrors to meet the schedule of assembly into the spacecraft, the XRCF was used to collect data with ACIS without any focussing optics. These data, collected during Phase I, fit naturally into the sub-assembly lab calibration data, and are discussed there.
For completeness, it is worth mentioning that data were collected at XRCF during September, 1996, using the ACIS-2C camera and the TMA (Test Mirror Assembly). Data from these tests are still available, but it is not anticipated that they will be heavily used in the ACIS calibration. The primary purpose of this testing was to serve as a rehearsal of the true AXAF calibration.
|Phase||Detector||Mirror||Start Date*||End Date|
|F||ACIS-2C||HRMA||12:11 22 Feb 1997||17:52 10 Mar 1997|
|G||ACIS-2C||HRMA||9:57 10 Apr 1997||8:41 12 Apr 1997|
|H||Flight||HRMA||23:56 17 Apr 1997||6:00 26 Apr 1997|
|I||Flight||none||6:45 7 May 1997||0:04 18 May 1997|
|R||ACIS-2C||TMA||7:00 30 Sep 1996||13:28 5 Oct 1996|
* Times are given in local (Central) time. Note these refer to the start and end of ACIS data collection intervals, not necessarily to the start and end of the XRCF Phase.