Next: Focal Plane Temperature Up: Effects of Focal Plane Previous: Effects of Focal Plane

### Detector Electronics Temperature

Prior to delivery to XRCF, the ACIS integrated flight instrument underwent testing at MIT Lincoln Laboratories. Some of these measurements involved operating ACIS with the Detector Electronics Assembly (DEA) at temperatures of -17 C and 20 C.These tests were used to map the gain dependence on the electronics temperature. A detailed description and analysis of the gain calibrations are reported in ACIS Memo #143 [Pivovaroff1997c]. The main points are summarized below:


-The DEA was stabilized at temperature Ti, (Ti = -17 C, 20  C)		 -Each chip was independently illuminated with an Al  source  and an 55Fe source
-The focal plane (FP) temperature during these measurements ranged between
-116--119 C
-Gains were calculated by fitting a linear relationship to a gaussian centroid for
the Al  and the Mn  line


Table 4.33 lists the mean gain for each chip for DEA temperatures of 20 C (hereafter LLTVhot) and -17 C(hereafter LLTVcold). Table 4.33 also lists the average chip gains calculated from the XRCF Flat Field (XRCFFF), Phase I data. See Section 4.3.1 for the details of these measurements.

 FPLocation Mean Gain (eV/ADU) LLTVcold XRCFFF LLTVhot DEA=-17 CFP=<-117.5> C DEA=13 CFP=-110 C DEA=20 CFP=<-117.5> I0 3.812 3.827 3.840 I1 3.809 3.849 3.857 I2 3.563 3.592 3.610 I3 3.839 3.882 3.891 S0 3.689 3.797 3.811 S1(BI) 4.405 4.413 4.455 S2 3.724 3.758 3.771 S3(BI) 4.492 4.507 4.574 S4 4.256 4.306 4.319 S5 4.075 4.101 4.119

It is clear from the data that the gains increase with increasing DEA temperature. In order to quantify this relationship, we have made a linear fit of the gains as a function of temperature. Figure 4.69 shows the linear best-fit plotted against the measured gain values. The hash marks simply indicate the data point and are not representative of errors. The LLTVcold data for I0 and S0 are known to have a problem with drifting biases, so these two data points are suspect. With only three data points to fit, the incorrect LLTVcold gains greatly influence the fits and skew the slope of the expected lines. Excluding I0 and S0, the data seem to follow the linear relationship quite reasonably. The small deviations from linearity are expected, as only two data were used to define the LLTV gains.

#### Footnotes

...suspect
This problem is fully characterized in ACIS Memo #143. The jitter.dacs software patch implemented during the XRCF Flat Field measurements eliminates this problem.

Next: Focal Plane Temperature Up: Effects of Focal Plane Previous: Effects of Focal Plane

Mark Bautz
11/20/1997